Determine the appropriate µV per graduation for your scale application.
Total load cell build.
Formula for Count By:
Count By = Scale Capacity / Graduations
mV/V * Excitation * 1000 * Count By(lb/kg)
(Total Load Cell Build)
This value should range from 1µV/graduation (or slightly less) to 30µV per graduation or more. Note: Always consult your digital weight indicator's specifications for the appropriate uV/graduation compatible with your device.